Insulin Pump control system
1 The Bolus Wizard calculator does not account for manual injections, and could prompt you to deliver more insulin than needed. Too much insulin may cause hypoglycemia. Consult with your healthcare professional for how long you need to wait after a manual injection before you can rely on the active insulin calculation of your Bolus Wizard.
2 Bergenstal RM, Tamborlane WV, Ahmann A, et al. Effectiveness of sensor-augmented insulin-pump therapy in type 1 diabetes. N Engl J Med. 2010;363:311–320.
3 Genuth S, Backlund JY, Bayless M, et al. Effects of prior intensive versus conventional therapy and history of glycemia on cardiac function in type 1 diabetes in the DCCT/EDIC. Diabetes. 2013;62(10):3561–3569.
4 Bergenstal RM, Klonoff DC, Garg SK, et al. Threshold-based insulin-pump interruption for reduction of hypoglycemia. N Engl J Med. 2013;369(3):224–232.
5 Ahmet A, Dagenais S, Barrowman NJ, Collins CJ, Lawson ML. Incidence of nocturnal hypoglycemia in pediatric type 1 diabetes: a pilot study using continuous glucose monitoring. J Pediatr. 2011;159:297–302.
6 Seven out of every 10 MiniMed 530G Threshold Suspend feature users feel that their worry of experiencing a low or a high during the night has been reduced.
7 User evaluations. Data on file, Medtronic MiniMed, Inc., Northridge, CA.
8 26, 053 patients kept Threshold Suspend (TS) function ON 100% of time and 4, 817 patients kept TS function OFF 100% of time. 8, 349 patients had time in both ON and OFF. Data from only the voluntary CareLink® Personal Uploads from MiniMed 530G in the U.S. available from October 15, 2013, to June 25, 2015 evaluated. BGs represent self-reported meter BGs in the pump.
9 dQ&A. (2015) dQ&A Patient Panel, Q1 2015, Summary Report. May 2015.
*Measured as sensor glucose levels.
**MiniMed Connect app user chooses who can access information.
MiniMed®, Enlite®, CareLink® and Bolus Wizard® are registered trademarks, SmartGuard™ is a trademark and StartRightSM is a service mark of Medtronic MiniMed, Inc.